Bandwidth of 0.1Hz -10 kHz allows users to record several types of extracellular potentials. • Low-cut filter (High pass filter) with 0.1, 1, 10, and 100Hz cutoff. • High cut filter (Low pass filter) with 1, 2.5, 5, 7.5, and 10kHz cutoff. • Built-in 2-channel stimulator allows users to program and deliver complex stimulation sequences independently through any 2 electrodes. • USB connection to computer.
Head Amplifier Number of channels 64 Gain x 10 Bandwidth 0.1Hz-100 kHz (+0dB to -3dB) Input impedance 100MΩ Output impedance 10kΩ
Stimulator Number of channels 2 Output format: Constant current Maximum input voltage +/-4V Maximum output current +/-200µA
Main Amplifier Number of channels 64 Gain 1-217 Bandwidth 0.1Hz-10 kHz (+0dB to -3dB) Analog high cut filter 1, 2.5, 5, 7.5, 10kHz (-6dB/oct) Analog low cut filter 0.1, 1, 10, 100Hz (-12dB/oct) Input impedance 100 MΩ
Digitizer Resolution 16 bit Sampling rate 20kHz Output: USB
Write a review
Sharing your experience will help scientists like you. Achieve Reviewer Status and Win an iPad 3 (All reviews published will be entered into the next drawing on January 30th 2014).